"I synthesized some 18A, an 18-amino acid amphipathic peptide ... The HPLC purity analysis indicates that the peptide is now >95% pure. Pretty good if you ask me! ... I then connected two identical cartridges inline through the luer fittings. I used the exact same gradient, as measured in column volumes, the same flow rate (no change in linear velocity), and injected the same crude peptide volume for the new “double” cartridge purification and selected run."
Find out what she learned in the post Can I improve my peptide purification by increasing the column length? on The Peptide Synthesis Blog ».
Finally - an Easy Urine Sample Preparation for LC/MS Analysis
A Smorgasbord of Analytical Methods
Extraction of Acidic Catecholamine Metabolites in Plasma Using EVOLUTE® EXPRESS ABN Prior to LC-MS/MS Analysis
Seminar: An Introduction to the Biotage Automated Workflow Solution
The 2018 Analytical Catalog is Here